# Owen’s Bridge – Phasor diagram & Advantages

Owen’s bridge is an AC bridge used for measurement of inductance of a coil in terms of capacitance. It is used for low Q-coils. Owen’s bridge is shown in figure 1 under balance condition.

Contents

From figure 1, there are four arms as

arm 1 = ab

arm 3 = bc

arm 4 = dc

where

L1 is an unknown self-inductance of resistance R1
R2 is a variable non-inductive resistance
R3 is a fixed non-inductive resistance
C2 is a variable standard capacitor
C4 is a fixed standard capacitor

Therefore, the impedances of arms 1, 2, 3 and 4 are respectively, At balance condition,

Z1Z4 = Z2Z3 On comparing real and imaginary part both side, we have  ## Phasor Diagram of Owen’s Bridge 1. The equations obtained are simple.

2. These equations do not contain any frequency part.

3. Wide range of inductance can be measured with this bridge.

1. The variable capacitor, C2 used is costly and has less accuracy.

2. The variable capacitor, Cvalue becomes large while measuring inductance for a high Q-coils.

This site uses Akismet to reduce spam. Learn how your comment data is processed.

error: Content is protected !!